CYCO pH 4.0 Solution 1 Liter (12/Cs)
Due to COVID-19 supply issues, we are operating only as a brick and mortar business. Stop in or give us a call for availability! 224-399-9877
The most common reasons for pH meter error is dry storage of the electrode and electrode contamination. These cause loss of accuracy, reduced performance and slower response times. Thoroughly clean the electrode after use and always store your pH meter with the cap firmly in place. Frequency of meter calibration is crucial and will optimize your pH meter’s performance. For calibration of pH meters at pH 4.0, pH buffer 4.0 @ 25°C (77°F) based on international (APHA) standard formula.
QUESTIONS & ANSWERS
Have a Question?
Be the first to ask a question about this.